Update the Shelly 3EM device to fw 1.12, where the functionality for neutral CT is supported. Then proceed with the following steps:
1. Lock N to phase A, don’t unlock A clamp.
2. Choose to calibrate N from A
3. If calibration fails, reverse N and try again.
4. When calibration is successful, move N clamp to Neutral wire.